Add:No.1, Huanzhen West Road, Fengchuan Town, Tonglu County, Hangzhou, Zhejiang, China
Gynaecology Instrument Gynecological ozone treatment device using low-temperature plasma technology and bio-engineering technology, the air in the air instantaneous decomposition of ozone and a large number of active molecules and other active ingredients, directly kill bacterial propagules, spores and fungi (kill rate of 99.9% Above), to promote the growth of newborn healthy cells.
1. Relevant units of ozone treatment (1) The current domestic ozone concentration is recommended to use ug / ml to express. (2) the volume of gas generated by the ozone generator Unit: ml. (3) Hour time of the ozone generator timing function: min. 2. Ozone output concentration (1) Ozone maximum output concentration of 60ug / ml. (2) Ozone concentration is adjusted to continuously adjustable, the concentration level
1. The relevant unit of the ozone treatment instrument
(1) the current domestic ozone concentration recommended ug / ml to use.
(2) the volume of gas generated by the ozone generator Unit: ml.
(3) Hour time of the ozone generator timing function: min.
2. Ozone output concentration
(1) Ozone maximum output concentration of 60ug / ml.
(2) Ozone concentration adjusted to continuously adjustable, the concentration of the order of 1ug / ml.
(3) Ozone apparent output concentration and the actual output concentration error ≤ ± 8%.
3. Ozone generator pipeline output pressure,
When the output pressure> 1 bar (1 bar = 10 Pa), prevent the occurrence of suction.
4. Ozone absorber
Gynaecology Instrument The ozone generator should have an ozone absorption function for the absorption and degradation of excess ozone.
5. Pipeline pressure detection and alarm system
Gynaecology Instrument The line pressure detection and alarm system is used to detect the pressure of the intake line, the internal piping, and the output line. It is used to detect and alert the airway pressure to be too large or too small, and to prevent excessive pressure and suction happened.
6. Programmable ozone generator self-test function
Self-test function is used to detect the equipment start-up status and work abnormal situation, to facilitate the medical staff to carry out medical work, to prevent medical accidents, and to facilitate maintenance personnel remote equipment maintenance.
- Biochemical Analyzer
- Collection, Conservation And Treatm...
- 2016 The Six Characteristics Of Med...
- Error Analysis Of The White Blood C...
- Spring Children's Health Care Knowl...
- What People Need To Check Blood-lipid
- Blood Transfusions Should Pay Atten...
- Silent Killer: Chronic Urate Nephro...
- Medical Device Industry Standards A...
- Domestic Developments In Blood Cell...
- Endoscopic Instrument Development B...
- Ligation Clips The Scope Of Applica...
- Polymer Ligation Clips Precautions
- Disposable Circumcision Anastomat T...
- Circumcision Anastomat Precautions
- Medical Equipment Medical Equipment...
- Gynaecology Instrument Working Prin...
- Medical Equipment Market Analysis
- Circumcision Anastomat The Principl...
- Endoscopic Instrument Flexion